UCI800:Setting up Biopack calibration


 * put a serious pack of paper into the printer :) - does anybody know how to make biopack save paper?
 * put, -labeled protein sample into the magnet. Lock, shim and tune the probe.
 * find tof and pw for pulse for  at tpwr=62
 * go to some experiment dedicated to biopack calibration or create a new one by typing
 * type the following or create a macro file and run it:

TROSY='n' gzlvl1=10000 gzlvl2=9600 mag_flg='n' np=1024 gain=30 tpwr=62 pwClvl=63 pwNlvl=62 at=0.048


 * enter the value for the proton pulse width pw=...
 * (in old school vnmr - we have vnmrj6.1D) click through menu buttons "Main menu">"Setup">"Protein">"Manage probe files">"Update probe file with these parameters"
 * wait till you see big yellow message on the screen something like "Probe file has been updated"
 * click buttons "Back">"Calibrate">"Full"

Questions
Does anybody know what is the point of doing "Update probe file with these parameters" before the calibration? And why enter tof and pw in advance when we're doing full calibration anyway?