UCI800:EPR calibrations

Field calibration
Current calibration file: 0401

Oct 9, 2008
Sample Cu(II) acetate

Data (double integral vs. concentration, M) 0.0001 2.7e6 0.0002 4.99e6 0.0003 7.35e6 0.0004 9.77e6 0.0005 1.24e7 0.0006 1.18e7 0.0008 1.87e7 0.0009 2.28e7 0.001 2.43e7

Slope: 2.4e10 M-1