User talk:128.200.203.33

On Bruker instruments the correct set-up sequence is:

1. define probe (first)

2. then read parameters (e.g. rpar PROTON all)

3. set the solvent (e.g. "solvent" command)

3. then "gpro". "rpar" only brings in generic parameters. "gpro" brings in calibrated pulse lengths ftom the defined probe.

That's true, thanks. The Bruker Manual is very likely needs more attention, too long etc. Please feel free to correct anything. I'll take a more careful look at it when I come back from the ENC. Looks like Graham is doing something on it too. I myself support a Varian system and run things on Bruker as a regular user - so perhaps I am not in the best position to work on those manuals, but I'll give it a shot. Evgeny Fadeev 21:29, 8 March 2008 (MST)